|Pump Type||Chiller Cooling||SKU||Price|
|Positive Displacement||Air Cooled||PS:6860P46A270D||
Designed for high heat removal in demanding environments, PolyScience DuraChill® Chillers provide robust and reliable temperature control for closed, external systems such as pilot plants, medical diagnostic equipment, metalworking lasers, and plastic molding machines. These chillers are suitable for most applications with their many options and accessories.Series Features
1.5 HP Models Feature
Please Note: Due to high demand, PolyScience DuraChill chillers generally have a 6-8 week lead time.
|Temp Range (°C)||5 to 35|
|MaxFlow Rate (l/min)||13.2|
|Temp Stability (°C)||±0.5°C|
|Digital or Mechanical||Digital|
|Media Racks Available||No|
|Compressor Size||1.5 HP||1.5 HP||1.5 HP||1.5 HP|
|Working Temp Range||5° to 35° C|
|Temperature Stability||±0.5° C|
|Electrical Req (VAC/Hz/Ph/A)||230/60/1/23.1||230/60/3/22||230/60/1/13||230/60/1/17.2|
|Pump Type||Turbine||Positive Displacement|
|Cooling Capacity @ +20°C||5200 W||5328 W||5200 W||5328 W|
|Max Pressure||100.00 psi|
|Max Pressure Flow||11.00 gpm||3.50 gpm|
Cooling capacity based on 20° C/68° F entering air and 20° C/68° F leaving water. Allowance made for heat gain from pump.
Can I run a DuraChill continuously under maximum load?
Under any situation requiring continuous operation, both we and PolyScience recommend that you don't run a chiller at more than 80% load. However, the 1.5 HP chiller over-performs its specifications somewhat, and could potentially be used continuously at maximum load so long as the operating conditions are favorable - essentially, that it is used in a clean, climate-controlled (not hot) environment.
What kind of coolant inlet / outlet connections do the 1.5 HP DuraChills have?
They have 1/2" female NPT connections. Two sets of Inlet/Outlet Adapters are included with purchase: ½ inch male NPT x ½ inch hose barb and ½ inch male NPT x 5/8 inch hose barb.
Do the DuraChill Chillers come on casters?
Yes, they all come with casters pre-installed.PolyScience 2016-01-17